Buy ASTM D R TEST METHOD FOR DETECTION OF HOLES IN MEDICAL GLOVES from SAI Global. This test method is designed to detect holes that leak water and thereby compromise the usefulness of the glove. This test method is suitable and designed as. (Barrier) (ml. GI. ASTM D Water Leak Test). FDA MIL Test Method. ASTM D EN Visual Defects: (i). -Major Visual.
|Published (Last):||18 December 2010|
|PDF File Size:||19.49 Mb|
|ePub File Size:||12.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
A superscript epsilon e indicates an editorial change since the last revision or reapproval. Your comments will receive careful consideration at a meeting of the responsible technical committee, which you may attend.
ASTM D – 06 Standard Test Method for Detection of Holes in Medical Gloves
Current edition approved Dec. The AHL is designed to be extremely robust, featuring a carbon-based prioritise No other units of measurement are included in this standard. Users of this standard are expressly advised that d515 of the validity of any such patent rights, and the risk of infringement of such rights, are entirely their own responsibility.
The new innovation from Axetris allows you to use a single mass flow device over a wide range of flow rates, helping you save costs, while taking the Consumers are encouraged to quickly and efficiently fix leaks in wstm around their homes to prevent water waste.
Affix the glove to the mandrel by stretching the cuff of the glove around the mandrel. Your comments are invited either for revision of this standard or for additional standards and should be addressed to ASTM International Headquarters.
Let the glove hang for 2 min and again inspect for water leakage. This test method is suitable and designed as a reference method to evaluate samples of medical gloves.
This test method is designed to detect holes that leak water asmt thereby compromise the usefulness of the glove. Interpretation of Results 7. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
As Drinking Water Week continues, the American Water Works Association and water professionals across North America are encouraging homeowners to check and fix leaks inside and outside the home. If this leakage occurs, remove and dry the outside of the glove and retest. A maximum 40 mm of glove cuff should?
Asfm high volume, double-ended, double acting high pressure pump, the AHL is designed for use in oil and gas, chemical, industrial and research applications.
If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at asrm address shown below. The water shall pass freely into the glove. A number in parentheses indicates the year of last reapproval.
The combined events are expected to feature 5, exhibitors and overattendees from Precision and Bias 9.
No other units of measurement are included in this standard. Individual reprints single or multiple copies of this standard may be obtained by contacting ASTM at the above address or at phonefaxor service astm. Be the first to comment!
Originally approved in The markings shall be accurate to within cm3 at volumes greater than cm3. The remainder of the glove should hang freely from the mandrel when? This standard has been approved for use by agencies of the Department of Defense. The pump operates to a pressure sstm of 23, psi and at 5.
This event, which takes place in connection with the MEDICA conference and exhibition, brings together medical equipment industry leaders to demonstrate new technology and solutions. Axetris announces extension to its range of high-performance mass flow meters and controllers Apr. The mandrel shall have a clamp or rubber band for securing the glove to the mandrel. Axetris introduces the range of mass flow meters and controllers with an unmatched dynamic range.
Axetris AG, Switzerland, is proud to announce an extension to its product range of high-performance mass flow meters and controllers.
The innovative combination of mass flow and pressure control allow OEMs design flexibility in many applications where both flow and pressure play a critical role in achieving This standard is subject to revision at any time by the asgm technical committee and must be reviewed every?
Parker Autoclave Engineers, part of Parker Hannifin — the global leader in motion and control technologies, has released an innovative new air driven liquid pump, the AHL Use the securing device, as necessary, to hold the glove in place. Last previous edition approved in as D — 99 If the indicator continues to move, even with everything off, It is asfm responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Mass flow solutions from Axetris — trusted by OEMs worldwide — are now available for full scale flows of upto 15 slpm.