Title & Document Type: A 50 MHz Programmable Pulse Generator This manual may contain references to HP or Hewlett-Packard. Hp A Pdf User Manuals. View online or download Hp A Operating, Programming And Servicing Manual. View and Download HP A operating, programming and servicing manual online. 50 MHz programmable. A Pulse Generator pdf manual download.
|Published (Last):||17 January 2006|
|PDF File Size:||16.61 Mb|
|ePub File Size:||9.80 Mb|
|Price:||Free* [*Free Regsitration Required]|
HP does not warrant that the operation of the instrument, software, or firmware will be uninterrupted or error free.
HP A Pulse Generator Operating & SERVICE Manual | eBay
This page was last modified on 29 Mayat Power Supply block diagram For example, STO 3 stores the current setting in location 3. EOI cannot easily be turned off. Any international shipping and import charges are paid in part to Pitney Bowes Inc. If a new plug is to be connected, it should meet local safety requirements and include the following features: The descriptions are covered in subsequent parts of this section.
Typical signals in External Width mode 1. Enables a data bus driver U36, which allows the remote programmer to read the address set on the HP-IB address switch. Detail of wiring to Line Voltage Selector switches Starting with the instrument in its Servicing position: Eight HP-IB data lines. Signature Analyzer Probe connections Marvin Test Solutions, a vertically-integrated aerospace test and measurement company, has created and delivered innovative, reliable test systems for factory, depot, intermediate, and flightline use since Press the [man] button 8.
See Chapter 2 Specifications for more details. The examples list the basic operating steps in the order in which they would normally occur after switching on.
Capacitance circuit This feature allows simple test sequence development as well as no need for a controller in repetitive tests, with reduced bus traffic in ATE applications. The parameter window will be automatically illuminated. In order to avoid influence from the control or main board, remove the device-bus cable from the main board A1 at connector J2 See Chapter Each vernier key increments or decrements the corresponding digit in the digital display. Product Description Price Qty.
The whole counter counts down when clocked by the burst-clock signal from the main board. A small part of the associated circuitry, concerned with burst control, is on the control board A2. ROM layout B B External Trigger Controls When the HP A is switched off, the 5.
This allows the power supplies to become established before the microprocessor starts running. Line Voltage and Fuse Selection 4- 1. Remove mamual screen covering the h; board.
Possible failures as in E Home Questions Tags Users Unanswered. Address Bus The address bus drivers and decoders can be checked using signature analysis. Normal output OUT, is passed via level shifter Q manuak the delay generator trigger input.
Hewlett Packard Operating & Service Manual for the A Pulse Generator | eBay
This manual applies directly to the instruments with the serial numbers quoted on the title page. U28 layout B B For instruments with higher serial numbers, refer to the Manual Change sheets in Appendix C Updating.
Duty Cycle Error There are three types of error which set the duty cycle error bit Bit 4 in the status byte. The hardware settling time must also be considered. The following supplies are then fed to voltage regulators, with potentiometers to adjust the final voltage level: Test Equipment 9- 1.
HP 8112A Manuals
See Details – opens in a new window or tab. Access to the HP Manjal for Adjustments Learn More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc.
Subroutine section The Servicc-srq subroutine. LOCAL The output signal and all instrument settings remain unchanged following a change from remote to local mode. Check the range decoder maanual Table Change HP A settings as follows: Normal output OUT, is passed via level shifter Q to the width generator trigger input.
Operation of the burst control circuit can be categorised as follows: The time for transferring Learn Strings may vary with different settings. This item may be a floor model or store return that has been used. Shaper IC Output on Pin 12 4.